Two orifice edge configurations are in general use in the United States. The square-edge type…
Orifice Plate Specifications
Orifice plate specifications are covered in API MPMS, Chapter 14.3/AGA Report No. 3 (1992) and include for example, the following:
• Plate thickness: The minimum and maximum thickness depend on plate size, but 1/8 inch is the minimum.
• Plate flatness tolerance: The maximum departure from flatness varies with orifice diameter (d) and size (D). It may be calculated as 0.005(D-d).
• Roughness: The orifice plate surface roughness should not exceed 50 microinches.
• Upstream edge: The square-edge (sharp-edge) type should not show a beam of light when checked with an orifice edge gage. Alternately, the square-edge type should not reflect a beam of light when reviewed without magnification. The orifice should not have a burred or feathered edge. Nicks on the edge can be expected to increase the measurement uncertainty.
• Centering: The concentric orifice should have the bore centered within 3% of the inside diameter of both the upstream and downstream sections of the orifice meter run.
• Beta ratio: A recent study by Chevron Petroleum Technology Company (CPTC) suggests that the optimal beta ratio is 0.4 to 0.55 for natural gas. Beta ratios departing from this range increase measurement uncertainty.
Categories: Flow Measurement | Comments Off on Orifice Plate Specifications